Sodium Iodide (NaI) Detectors
Sodium Iodide is a type of detector within a class known as scintillation detectors. Gamma rays penetrate through the detector housing and into the NaI crystal where they may experience interactions with the atoms inside the crystal, as shown in Figure 1.
These interactions can cause photons of visible light to be released from the atom and travel short distances within the crystal. Thallium (Tl) is used as a dopant for NaI because it helps to increase the number of visible photons being output by the crystal and also helps to improve the crystal's transparency to those emitted photons so they can be measured by equipment outside the crystal. As shown in Figure 2 these photons of visible light are detected by a photocathode to produce an electrical signal, which is then amplified and processed to create a gamma ray spectrum.
Figure 2 - Sodium Iodide detector assembly. Visible light is created inside the NaI crystal due to incoming gamma rays. This visible light is converted to an electrical signal by the photocathode. This electrical signal is amplified by the Photomultiplier Tube (PMT) and then processed into a spectrum by the Multichannel Analyser (MCA).
Report a problem or mistake on this page
- Date modified: